Understanding Leakage Inductance
An explanation of Leakage Inductance in transformers, why it is important, and how to best perform measurements
1, What Is Leakage Inductance?
Leakage inductance is an inductive component present in a transformer that results from the imperfect magnetic linking of one winding to another.
Any magnetic flux that does not link the primary winding to the secondary winding acts as inductive impedance in series with the primary, therefore this "leakage inductance" is shown on a schematic diagram as an additional inductance before the primary of an ideal transformer.
In certain applications, such as switchedmode power supplies and lighting ballasts, leakage inductance of the transformer may play a critical function in the product design. For this reason, accurate measurement of leakage inductance is often an important test function for transformer manufacturers.
In order to avoid confusion with other transformer characteristics, this technical note will not refer to other components of loss such as winding resistance or interwinding capacitance.
Ideal transformer
For a theoretical, ideal transformer, there are no losses. Voltages are transformed in the direct ratio of the turns; currents in the inverse ratio of turns (figure 1).
Real transformer
In a real transformer, some of the flux in the primary may not link the secondary winding.
This "leakage" flux takes no part in the transformer action and can be represented as an additional inductive impedance that is in series with the primary winding (figure 2).
Real transformer plus an air gap
In certain transformer designs, leakage inductance must be a greater proportion of the total inductance and is specified within a tight tolerance.
The increased proportion of leakage inductance is usually achieved by introducing an air gap in the core design, thus reducing the permeability of the core and therefore the value of primary inductance.
The ratio of flux that does not link the primary winding to the secondary winding will therefore increase relative to the flux that links both windings (figure 3).
2, Why is measurement of leakage inductance important?
Leakage inductance (LL) may be undesirable in a wound component, in which case it is important to measure the value to show that it is low or, in some applications, such as electronic lighting ballasts and resonant power converters, leakage inductance is deliberately introduced and its value is an integral part of the circuit design.
In these applications, the leakage inductance provides an energy storage medium that is essential to achieve correct operation of the finished product.
It is therefore important that the value of leakage inductance of the transformer is known to be within specified limits.
3, How is leakage inductance measured?
When an LCR meter is connected to the primary winding of a transformer with opencircuit secondary terminals (figure 4), the value of inductance (L) comprises primary inductance (LP) plus leakage inductance (LL).
Since LL is a function within the transformer, it is clearly not possible to measure its value directly.
A method must therefore be used to subtract the value of LP from the total measured inductance.
This is achieved by applying a short circuit across the secondary terminals (figure 5).
A perfect short circuit will result in zero volts on the output terminals (figure 6) and, through transformer action, zero volts will also appear across the primary inductance.
The measured value of inductance at the primary terminals will therefore be the true leakage inductance (LL).
Unfortunately, achieving a perfect short circuit on the secondary of a transformer is difficult in a laboratory and completely impractical in a production environment.
In production, it is common for the short circuit to be applied manually or via a switchable relay.
Under these conditions, a perfect short circuit cannot be achieved, and it follows that the secondary voltage will not be truly zero.
The voltage attributable to the imperfect short circuit will then appear across the primary inductance as a shortcircuit error multiplied by the turns ratio (figure 7).
Ls/c is reflected in the primary as N^{2}Ls/c because, in any winding, L is proportional to the number of turns squared (L α N^{2}).
Thus, Ls/c is reflected as a function of:
( Np / Ns ) ^ 2 = ( Lp / Ls )
The measured value of the primary inductance can be considered vectorially as the sum of the leakage impedance plus the reflected impedance of the shortcircuit error. This is shown in figure 8.
4, The traditional solution
In order to obtain the true value of leakage inductance, engineers will carefully apply a soldered short circuit to the secondary of the transformer to be tested and measure the value of inductance on the primary.
This value of inductance will be recorded as the 'true' leakage inductance (e.g. 150μH).
The inductance will then be measured on the same transformer after the soldered short circuit has been replaced by either a shorting clip or a fixture with relayoperated short circuit, depending on the technique that will be chosen for production.
The measured inductance is again recorded (e.g. 180μH).
This value will, of course, be greater than the original because it includes the true leakage inductance plus the shortcircuit error inductance.
The difference between these two values (in our example 30μH) is then used in production testing as a fixed offset that is programmed into a production LCR meter to obtain an approximation of the correct value in the presence of an imperfect short circuit.
In practice, it is impossible to achieve a relaybased or manualbased short circuit that produces exactly the same shortcircuit error every time.
This nonrepeatability of shortcircuit error is such that the fixed offset cannot provide a production department with accurate and repeatable results.
This is illustrated in the following table:
True LL 
Meas. value 
Fixed offset 
Result 
Pass/fail 

Meas. #1  150 μH  180 μH  30μH  150 μH  ✓ 
Meas. #2  150 μH  200 μH  30μH  170 μH  X 
Meas. #3  150 μH  250 μH  30μH  17 5μH  X 
5, The Voltech solution
Voltech have developed their AT series testers with an architecture and processing ability to remove the shortcircuit error from the primary inductance measurement during each and every test.
A simplified version of this process is shown below.
Firstly, as part of the LL test, a silent measurement is made of the Turns ratio of the part under test.
The Voltage on the secondary is also measured when the secondary is open circuit.
This gives us Vopen on the above graph
Second, With the (non ideal) short applied to the secondary, the voltage and current are also measured.
This gives us the point V1/I1 on the graph.
These two points are then extrapolated (on the assumed linear V/I line) an calculated back to where V=0 to give Ishort.
This is the expected current flow in the secondary under the Ideal short condition  i.e. where the short is perfect and there is no voltage drop across the secondary.
This Ishort value combined with the earlier silent TR result can be used to calculate the corresponding current effect on the primary side, and hence removed from the LL result measured on the primary.
This is a simplified venison of the technique.
In reality, the measurements are a combination of real and imaginary measurements, so the technique is shown vectorially below
From the primary vector diagram, it can be seen that each measurement is the sum of the voltage attributable to the leakage inductance plus the error voltage from the secondary short circuit.
Before applying a short circuit, the Voltech AT series testers measure the primary to secondary turns ratio.
The testers then automatically apply a short circuit, using an internal relay matrix, and measure the shortcircuit voltage at the transformer secondary pins.
The vector of this short circuit voltage is automatically multiplied by the turns ratio, producing an 'error vector' that is equal to the shortcircuit error voltage reflected into the primary measurement.
The leakage inductance is then computed from the total primary inductance value less the primary error vector that has been calculated.
This process enables Voltech AT series testers to provide the true leakage inductance value, irrespective of short circuit variability.
True LL 
Meas. value 
Realtime vector comp. 
Result 
Pass/fail 

Meas. #1  150 μH  180 μH  ✓  150 μH  ✓ 
Meas. #2  150 μH  200 μH  ✓  150 μH  ✓ 
Meas. #3  150 μH  250 μH  ✓  150 μH  ✓ 
6, Leakage Inductance Conclusion
Leakage inductance is a critical transformer characteristic that presents a particular measurement challenge to both design and production test engineers.
By looking at the factors effecting measurement integrity and developing innovative measurement techniques to overcome these factors, Voltech provides a unique solution to a problem of measurement variability that faces almost all transformer manufacturers.
Should you have questions on any of the other test functions available for the Voltech AT series transformer testers, please do not hesitate to contact us.