PM100CE / PM300CE FAQ
Serial numbers PM100CE (AQxx/xxxx and AXxx/xxxx) and PM300CE(Serials ARxx/xxxx and AYxx/xxxx)
How do I update firmware? What is the latest firmware?
The latest PM300CE firmware is V2.09
The latest PM100CE firmware is V2.10
Firmware is not user upgradeable.
Where is the user manual for the Unit / VPAS Lite / RS232 /IEEE / Alarm card?
All user manuals are linked below
How do I Calibrate the unit?
Any good calibration house should be able to calibrate &verify the unit to the spec.
The Calibration method is contained within the user manual
TIP on manual calibrations using front panel and your own source.
The lower ranges will say something like “Enter 120mV”, but the firmware actually expects you to enter “0.1203” or what ever your REAL applied is, in VOLTS.
The prompt is misleading so users enter this as per the prompt – i.e. “120.3” in mV, and the cal fails.
You should find the same issue on the A / mA ranges too
Is the VPAS Lite software still available?
The VPAS Lite software for data logging is no longer available for new sales. the last version is V2.08
The software is coded to work only for an individual unit serial number.
User Manual below