Can the AT3600 test 3-phase transformers?
There is no particular 3-phase interface for the AT3600, but many customers do test three-phase transformers.
If the transformer is wound with each phase on a separate core, then each primary and secondary combination can be tested - in one program - as three separate transformers. Resistance - Magnetizing Current - Voltage Open Circuit - Watts, Surge and HiPot etc. can be applied as normal.
This will check:
Resistance - correct wire, properly terminated
Magi - correct number of turns, on correctly assembled core material
Voltage Open Circuit - Correct number of turns wound and terminated in the right direction
Watts - correct number of turns, on correctly assembled core material
Surg - Good wire insulation, no dropped turns
Hi-Pot - Good insulation between windings and to core