How are the 2 inputs in the user port used. The pinouts and electrical requirements are shown but not how to read them. Can they be used to pause the testing at a certain point?
You cannot use the input of the user port for this but we do have a method for pausing as follows:
There is no PAUSE step in the program. However, we provide a method to realize this PAUSE function as below.
If you intentionally make one of the test steps FAIL and then you insert a TRIM test (Trimming Adjustment Test) immediately after that step, it will make the test program PAUSE to allow the user to make some adjustments.
You also have to “fix” that previous FAILED test step during this PAUSE. Then you can press the "CONTINUE" button to make the program continue to run. The program will continue from the previous FAILED step to the end.
Information for how to add TRIM Test (Trimming Adjustment Test) is on Page 3.4.39 of the "Voltech AT3600 User Manual"