Can the following be tested in one program: VOCX: Primary: 3A / 50Hz, secondary, also Primary : 30A / 50Hz, secondary 0.5V. In addition IR and ACHP?
1. The VOCX test (using the AC interface module) controls a voltage source, to supply 3A of current it will be necessary to set a voltage (0-600V) with a series resistance to supply 3A
It should be possible to support the measured voltage
2. The AC interface module cannot support above 10A.
3. IR and ACHP will not be a problem if tests are within available test parameters.