02 APR 2020
ZB - Fix to ZB Compile process to deal with STAT 0020.
HPDC - Fixed error code 0020 on IR and DC-EHT following a MAGI test. The issue was caused by the trip system latching during the shutdown process and not clearing for the HPDC test.
DCRT - Fixed DC ramp test reporting pass when it should fail.
LL - Fixed LL accuracy issue caused by running open circuit compensation.
LSB LPB ZB - Added Offset measurements to the LPG, voltage channel and current channel to improve stability of LSB, LPB and ZB measurements.
R - Improved compensation model for the resistance test.
TRL - Fixed TRL trimming issue on secondary caused by failing to preserve the shunt for the polarity measurement.
SELF TEST - ID1020 now has the filter state set properly so does not report false fails.
SELF TEST - Adjusted the self-test limits for GN01 open and close timing to reduce false self-test failures
KNOWN ISSUES - Issues being worked on for future release
Improvements to capacitance accuracy at very low capacitance <10 pf
Fails DCRT - occasional status errors or mis-reads - Suggest HPDC as alternative.
Fails ACVB - occasional status errors or mis-reads. Suggest HPAC as alternative.
Fails DCVB - occasional status errors or mis-reads. Suggest using HPDC as alternative
HPDC - Occasional Status errors and false fails when close to measurement limit.
FUTURE DEVELOPMENT - COMING SOON
FRONT PANEL MEASURE MODE - run individual tests from the front panel
CHINESE language - Translation of all front panel menu text.