3.6.2 AT Compensation with Multiple DC1000s
When compensating your Voltech AT product for use with the DC1000(s):
- Measure the open circuit inductance without the DC1000s connected, by performing “AT measure” with the Editor. Note this inductance value.
- Connect your DC1000(s) to the device under test, with all output turned on, and current set to zero.
- Subtract the inductance value now from the inductance value taken without the DC1000(s) connected. Use this value as the negative offset when running an AT program.